Background: Vascular closure devices are routinely used following femoral artery access to perform percutaneous coronary interventions (PCI). A number of devices are available on the market. We have reported previously on our experience of the Mynx device following diagnostic coronary angiography.
Aims: To assess the success and complication rates of the Mynx device used in all-comers in the context of PCI within a single cardiac centre.
Methods: Retrospective analysis of data available for patients who underwent PCI via the femoral route and received a Mynx device at a single tertiary centre.
Results: The device was used to achieve haemostasis in 113 patients following PCI. In all cases weight-adjusted Heparin as well as dual antiplatelet therapy (Aspirin and Clopidogrel/ Ticagrelor) was administered as per PCI protocol. The device was successfully deployed in 111 cases (98.2%). The were 2 cases of device failure, one due to operator error and the other due to Mynx grip balloon bursting during device deployment. In 15 cases (13.2%; 9 male and 6 female) there were reports of small haematomas (<2cm) or oozing resulting in application of manual pressure or Femstop (St. Jude). A further patient required ultrasound-guided compression of the artery due to a large retroperitoneal bleed resulting from access complications. A larger proportion of the cases with complications were done as PPCI (44% vs 33%). The group with complications had higher systolic BP (140mmHg vs 128mmHg; P<0.05) and MABP (97mmHg vs 75mmHg; P<0.05) as compared with the group without any complications.
Conclusion: The Mynx closure device is safe and easy to use in the context of PCI, in both the elective and emergency (PPCI) settings. Complication rates, predominately minor, can be minimised in experienced hands.
Background: Vascular closure devices (VCD) are routinely used to achieve haemostasis following percutaneous arterial procedures. The extravascular polyethylene-glycol based MynxGrip® device (Cardinal Health) received FDA approval for use in the closure of femoral veins, but so far limited data is available on its use, especially with concomitant use of anticoagulants.
Method: This is a retrospective analysis of data from a single-centre on the effectiveness and complication rates following the use of the MynxGrip® device for femoral venous closure in patients undergoing diagnostic/interventional (temporary pacing during balloon aortic valvuloplasty, or electrophysiology) procedures utilising 5-7F sheaths.
Results: 85 patients (mean age 74 years) underwent femoral venous closure with the MynxGrip® device. 51.8% were male. The rate of concomitant anticoagulant or antiplatelet use was 52.9%. Device deployment was 100% successful with full haemostasis in all cases. There were no major vascular complications (bleeding, thrombosis, or infections). There was one case of a minor small venous hematoma which did not require treatment. The mean length of stay was less than 1 day (67.1% patients discharged the same day) and overnight stay only indicated by interventional procedure.
Conclusion: These data support safety and efficacy of the MynxGrip® device for femoral venous closure with same-day discharge, even with concomitant aggressive antiplatelet and anticoagulant use. It has the potential for use in other large bore venous access sites.
My candid opinion is that the service you render is second to none. My favourite part is the prompt response to issue, really i value that.
Abiodun Akanbi Adeogun
Dear colleagues! I am satisfied with our cooperation with you. Your service is at a high level. I hope for a future relationship. Let me know if I can get a paper version of the magazine with my articles from you. I see them on the Internet.
The service from the journal staff has been excellent.
We thank to the heighten science family, who speed up the publication of our article and provide every support.
Great, We are too comfortable with the process including the peer review process and quality. But, the journal should be indexed in different databases such scopus.
“Mobile apps and wearable technology are becoming ubiquitous in our environment. Their integration with healthcare delivery is just beginning to take shape. The early results are promising and the possibilities great."
BS, PharmD., MBA, CPHIMS, FHIMSS, Adjunct Professor, Global Healthcare Management, MCPHS University, Chief Strategy Offi cer, MedicaSoft, Senior Advisor, National Health IT (NHIT) Collaborative for Underserved, New York HIMSS, National Liaison, Health 2.0 Boston, Past Chair, Chair Innovation, USA
I wanna to thank Clinical Journal of Nursing Care and Practice for its effort to review and publish my manuscript. This is reputable journal. Thank you!
The submission is very easy and the time from submission to response from the reviewers is short. Correspondence with the journal is nice and rapid.
Once I submitted the manuscript, the response time of the reviewers was very fast. The fine-tuning of the galley proof was likewise prompt. I believe the journal provide a valuable outlet to disseminate physical rehabilitation scientific knowledge to the clinical community.
I wanna to thank clinical journal of nursing care and practice for its effort to review and publish my manuscript. This is reputable journal. Thank you!